Pursat Province Pursat Province Referral Hospital (PRH)
Kampong Chhnang Kampong Chhnang Province Referral Hospital (PRH)
Kampong Cham Kampong Cham Province Referral Hospital (PRH)
Kampong Thom Kampong Thom Province Referral Hospital (PRH)
Kampong Thom BaraySantouk Referral Hospital (RH)
Kampong Speu Kampong Speu Province Referral Hospital (PRH)
Kep Kep Province Referral Hospital (PRH)
OddorMeanchey OddorMeanchey Province Referral Hospital (PRH)
Specifically designed for red bag Medical wastes with high density and high moisture content.
Apart from the manual loading of waste, the incinerator should be automatic and easy to use or Plug and play modal
The incinerators shall be able to be operated with the following three forms of fuel and gas: Diesel, LPG and N-Gas.
Maximum load capacity of a minimum of 100kg
Burn Rate of anywhere between 50 – 80 kg/hr
Thermal Capacity 190 kwg/hr or less
It should use a Power Supply of 50/60Hz or 210/230v
External Length of anywhere between 1800 to 1900 (mm)
External Width of anywhere between 1100 to 1200 (mm)
External Height of anywhere between 1900 to 2000 (mm)
10 – 15mm Steel casing
high grade paint finish
Light weight resilient refractory fiber insulation door for thermal efficiency
It should have a Temperature indication and control center for each chamber
It should have a Fuel tank of capacity 200 liters ±20 liters
It should have a chimney stack of 6 to 8 meters.
6mtrs will be used in the rural areas while 8mtrs will be used in the areas within the provincial town
A minimum of 2 chambers
The walls of each chamber should be constructed with high grade refractory brick of a thickness anywhere between 210 – 240mm
Each chamber should have combustion burner air fans which are automatically controlled to ensure efficient heat distribution
Each chamber should have at least one temperature sensor
One temperature sensor mounting point at the base of the exit flue ensuring that chamber reaches the necessary 800°C minimum.
Minimum 50-100% moisture
Blood, drips, human faces
Majority of Cancer treatment wastes
Majority of Pharmaceuticals
2 years warranty on parts and services.
The supplier shall be able to provide and replace all below mentioned major spare parts for a period of 5 years (from delivery), at their own cost.
Primary & Secondary Chamber Thermocouple
Burner Ionization Probes
Burner Ionization Leads
Burner Flame Control Box
Burner ignition transformer
Complete spare burner
The aim of the training is to enable the participating local users/ technicians to use, maintain, repair and install incinerators.